Table of Contents
- 1 How is the precision of a dial comparator determined?
- 2 How accurate is a dial gauge?
- 3 What are the two basic types of indicator movements?
- 4 What is the working principle of dial test indicator?
- 5 How do you read a .001 dial indicator?
- 6 What is the principle of dial indicator?
- 7 How often should dial indicators be calibrated?
- 8 Which is better dial comparator or one Rev indicator?
- 9 Which is comparator is best suited for inspection of?
- 10 Why are dial indicators not a high resolution readout?
How is the precision of a dial comparator determined?
Precise quality of the gear forms and bearing freedom determines the repeatable precision of measurement achieved. Dial indicators typically measure ranges from 0.25 mm to 300 mm (0.015in to 12.0in), with graduations of 0.001 mm to 0.01 mm (metric) or 0.00005in to 0.001in (imperial/customary).
How accurate is a dial gauge?
A dial gauge is also used for measuring concentricity of round objects. These are available in inches as well as in millimetre. Inch dial gauges of 0.001 “measuring accuracy is in general use, but gauges are available up to the accuracy of 0.0001”. The commonly used metric dial gauge has an accuracy of 0.01 mm.
Do dial indicators need calibration?
Dial indicators help users measure the distance between two plates, but the readings given by the tool could be faulty. Calibrating a dial indicator prior to using it is the only way to be sure that the tool is reading correctly and its measurements are accurate.
What are the two basic types of indicator movements?
In the most general sense, there are two types of indicators commonly used in shop environments: test indicators and drop indicators.
What is the working principle of dial test indicator?
The working principle of the Dial Indicators is that the movement of the spindle multiplied with the reading of the main scale due to the indication of the needle. In addition to this, at the main scale rotation of pinions and gears is indicated.
Which of the following is attached to dial face for?
Explanation: Tolerance pointers can be attached to the dial face for rapid checking of the parts. These pointers can be used to clearly indicate any tolerance, whether bilateral or unilateral, on the dial gauge.
How do you read a .001 dial indicator?
How to Read A 0.001 Dial Indicator
- multiply the total inner scale divisions to 0.1″ (for example, 2 divisions, then 2 x 0.1 = 0.2″)
- multiply the total outer scale divisions to 0.001″ (for example, 10 divisions, then 10 x 0.001″ = 0.01″)
- add those two values of inner scale and outer scale (0.2″ + 0.01″ = 0.21″)
What is the principle of dial indicator?
How do I calibrate my dial?
To calibrate you need a gauge stand, a surface plate, and appropriately sized gage blocks. The dial indicator is mounted on a gage stand on the surface plate, a zero is set on the dial, then gage blocks are inserted, and dial indicator reading is taken to determine that it is functioning properly.
How often should dial indicators be calibrated?
4.1 Accurate linear measurements are required to obtain proper data for laboratory and field use. 4.2 Calibrate dial indicators when purchased and annually thereafter.
Which is better dial comparator or one Rev indicator?
A dial comparator is very much like a one rev indicator. The part has to fall within the measuring range of the indicator or else it shows up as being off scale. There is no chance of being read incorrectly. So what’s the bottom line? A dial comparator has five to ten times better resolution and accuracy than a dial indicator.
Which is the lowest priced dial measurement tool?
The lowest priced measurement tool is, of course, the dial indicator. But dial indicators are not normally thought of as providing a high resolution readout. Manufacturing dial indicators to the highest possible standards will result in only modest improvement of resolution.
Which is comparator is best suited for inspection of?
Precision measurement of angles of non-mirrorquality surfaces is possible only with (a) a laser autocollimator (b) a visual autocollimator (c) an angle dekkor (d) any of these:comparatorsh6 C 1.
Why are dial indicators not a high resolution readout?
But dial indicators are not normally thought of as providing a high resolution readout. Manufacturing dial indicators to the highest possible standards will result in only modest improvement of resolution. This is because the substantial number of parts in a dial indicator generate a build-up of tolerances.